ST 300 SPIDER Line Analyzer, detection of eavesdropping devices
|Power supply||rechargeable battery|
|Modifications||In stock in||Qty||Buy|
Wired communications analyzer ST 300 SPIDER is designed to detect and locate eavesdropping devices, galvanically connected to power and low-current wire lines in the inspected object. The analyzer uses both passive and active modes of operations. This allows detecting eavesdropping devices which are in active or in stand-by mode at the time of a sweep operation.
These functions include:
• Detection and evaluation of low-frequency signals in low-current conductive lines;
• Activation of electret cable microphones by filing an in-line bias voltage;
• Detection of eavesdropping devices signals, which are transmitting information via power and low-current lines in the frequency range from 100 kHz to 150 MHz;
• Detection and assessment of unauthorized galvanic connections to conductive lines using methods of non-linear junction detection and reflectometry;
• Tracking wire lines laid in walls and other building structures using trace locator.
Use of automated mode of analyzer, in combination with electronic switch, allows to carry out various kinds of measurements on all the possible combinations of pairs of multi-wire cable in a few seconds. Adapters, couplers and cables, which are included in the set of ST 300 SPIDER delivery set, allows to connect the device to the most common types of wired lines. The device can be used in standalone mode or controlled from a personal computer.
The solution of different ways to detect wired eavesdropping devices is ensured by the multifunctionality of ST 300, appropriate accessory pack and the use of passive and active modes of operation.
Systems technical and programming basis, implemented in the structure and unique algorithms allows to use it in the following modes:
• Electronic switch;
• Low frequency amplifier;
• Wireline receiver;
• Nonlinear junction detector for wire lines;
• Trace locator.
Switching ST 300 in any of these modes is carried out compulsorily as at the same time the device can only operate in one of the main modes.
|Low frequency Amplifier|
|Frequency range, Hz||20....25 000|
|Input impedance, Ohm||200|
|Amplification regulation range, dB||0....60|
|Maximum signal amplitude on the entrance, V||+/-13,8|
|Spectral density of voltage noise nV,Hz||3|
|Bias voltage range, V||+/-26|
|Signal presentation form||oscillograph, spectrogram|
|Frequency range, kHz||100....150 000|
|Time to scan the entire range, S||2|
|The minimum detectable signal level, dBm||-50....-75|
|Dynamic range, dB||55|
|Input impedance, Ohm||50|
|The filter bandwidth, kHz||80|
|The maximum allowable voltage In||250(AC), 80(DС)|
|The presentation of information||spectrogram, oscillograph, table|
Non linear junction detector
|Signal Amplitude, V||0....13|
|Signal frequency, Hz||45....55|
|Split indication of even and odd harmonicas||present|
|Minimally detected level of non linear distortions, dB||-60|
|Diapason of distance, m||0...100|
|Measurement error percentage, %||+/-1|
|Possibility of operation in lines with voltage||no|
|Measurement range DC, V||+/-80|
|The measurement range AC, V||13|
|Frequency range, Hz||0....30 000|
|Carrier frequency of the test signal, kHz||100|
|Frequency of modulation of the signal, kHz||1...2|
|Signal amplitude, V||13|
|Possibility of the contact connection to the line||Yes|
|Built-in Li-pol accumulator with voltage, V||3,7|
|Consumed power, V||1,2....2,5|
|Time of continuous operation at maximum consumed power, hour||>7|
|Charging time of fully charged accumulator, hour||7|
|Dimensions of the main unit (length, width, height), mm||175х83х36|
|Weight of the main unit, kg||0,375|
|Package dimensions (length, width, height), mm||390х310х170|
|Weight of the set in the package, kg||4,4|
|Power supply||rechargeable battery|
|Indication of operation||light, sound|
|Battery life||at least 7 hours|
|Battery charging time||7 hours|
Delivery set:Delivery set:
1 – USB cable for connecting ST300 to a laptop - 1 pc.
2 – cable for telephone junction-box testing - 1 pc.
3 – cable for connecting the electronic switch to RJ45 sockets - 1 pc.
4 – cable for connecting the electronic switch to twin wire low voltage lines - 1 pc.
5 – probe for the cable locator receiver - 1 pc.
6 – cable for testing power mains equipped with electric sockets - 1 pc.
7 – tester screwdriver - 1 pc.
8 – cable for connecting the electronic switch to telephone sockets - 1 pc.
9 – charger - 1 pc.
10 – ST300 main unit - 1 pc.
11 – flashcard with software and documentation - 1 pc.
12 – crocodile clamp - 2 pcs.
13 – mains adaptor - 1 pc.
14 – cable for the test sound player - 1 pc.
15 – test sound player - 1 pc.
16 – electronic switch - 1 pc.
17 – receiver of the cable locator - 1 pc.
18 – accumulator batteries for the cable locator receiver (ААА) - 2 pcs.
19 – headphones - 1 pc.
20 – pass-through socket (RJ45) - 1 pc.
21 – splitter (RJ45) - 1 pc.
22 – splitter (RJ12) - 1 pc.
23 – short-circuiter socket (RJ45) - 1 pc.
24 – telescopic arm of the cable locator receiver - 1 pc.
25 – multiwire cable connector (with a screwdriver) - 1 pc.
26 – ground connection cable - 1 pc.
27 – device passport - 1 pc.
28 – protective case with custom panels - 1 pc.
Please note: software upgrade can cause recorder’s breakdown. This is not a warranty case. All the new recorders have the latest software version and there is no necessity to upgrade software after buying the recorder. Don’t take the risk. In case there are problems with audio recorder operation please apply to our technical support specialist: email@example.com